# Supercapacitors(EDLC) FAQ

Characteristics

How to calculate capacitance and ESR value?

Murata's supercapcitor(EDLC) is a suitable product for using with relatively large current or high power, so we measure nominal capacitance at 100mA.

Calculation of capacitance <Discharge method>

 Temperature: 25°C+/-5°C Discharge supercapacitor after charging by max voltage for 30 minutes according to the profile and circuit (see figure 1) . Id (A) =Discharge current (Constant) = 0.1A Figure 1 Charge/discharge current: 100mA V80%: 80% of Max voltage V40%: 40% of Max voltage t1: time to V80% t2: time to V40% Id: Discharge current  (Constant) Capacitance is calculated by the following formula (1).    (1) measuring circuit

*Reference: V80%-V40% in case of capacitance at 100mA discharge

 100mA 100% Charge/discharge current 1A 10mA 1mA 0.1mA Capacitance (Consider capacitance at 100mA discharge as 100%) 95% 103% 107% 116%
 Nominal capacitance

Calculation of ESR <AC method>

ESR is measured by AC method. It is calculated with the following formula (2) by measuring voltage of both sides of the capacitor (Vc) applying 10mA

Temperature: 25°C+/-5°C 1kHz 10mA Vc

We appreciate your cooperation with the FAQ improvement questionnaire.
Were these FAQs helpful?

We would like to hear your opinions and requests regarding these FAQs.
Opinions from customers will be used to improve the FAQs.
We will not respond to inquiries and requests received by this form.

Send inquiries from the following form, for problems that cannot be solved by FAQs and require a response from Murata.